Reverse engineering of the Fujifilm PMT12A PMT module that contains the photomultiplier tube, high voltage power supply and analog amplifier from a Fujifilm FCR XG-1 x-ray image scanner / digitizer.
Also known as Computerized Radiography (CR), it fulfills the job of scanning a x-ray exposed image plate with a laser beam and extract each pixel illumination with a photo multiplier tube setup. It can scan 72 plate per hour with a maximum resolution of 7080 X 9480 in 12 bits per pixel gray scale.
Pictures and long description: https://kaizerpowerelectronics.dk/high-voltage/fujifilm-photomultiplier-tube/
Discussion of reverse engineering: https://highvoltageforum.net/index.php?topic=1618.0
Discussion of the part 1 and 2 teardown: https://highvoltageforum.net/index.php?topic=1584.0
Part 1, Video: https://www.youtube.com/watch?v=G5EP17UfxKM
Part 1, Pictures and long description: https://kaizerpowerelectronics.dk/teardown/fujifilm-fcr-xg-1-x-ray-image-scanner-teardown-part-1-of-3/
Part 2, Video: https://www.youtube.com/watch?v=fL_aKOFaXbw
Part 2, Pictures and long description: https://kaizerpowerelectronics.dk/teardown/fujifilm-fcr-xg-1-x-ray-laser-module-teardown-part-2-of-3/